Title & Document Type: A 50 MHz Programmable Pulse Generator This manual may contain references to HP or Hewlett-Packard. Hp A Pdf User Manuals. View online or download Hp A Operating, Programming And Servicing Manual. View and Download HP A operating, programming and servicing manual online. 50 MHz programmable. A Pulse Generator pdf manual download.
|Published (Last):||18 May 2015|
|PDF File Size:||11.63 Mb|
|ePub File Size:||11.8 Mb|
|Price:||Free* [*Free Regsitration Required]|
Reference current input levels Triggering can also be simulated using the MAN key.
HP does not warrant that the operation of the instrument, software, or firmware will be uninterrupted or error free. They are not used in programming messages.
HP Hewlett Packard 50 MHz Pulse Generator 8112A
Instrument Overview This chapter deals with the overall instrument and is intended to help you to isolate a fault at a functional level. Refer to Chapter 4 Operating for additional information on parameter adjustment. Press the SET I key 3. Check the input signal from the delay generator at TP2 pin 7 of U against Figure Below is up screen dump of the results printed when running the program.
Inverts the output signal when lit. If the instrument battery has failed, the Standard Parameter Set is selected. Trending Price New. PER 1 fis Mode E. One pair of pulses manal each period up to 25 MHz mqnual rate. The parameter window will be automatically illuminated. Remove the rear of the instrument by unscrewing the two TORX fastening screws. The HP office will arrange for repair or replacement without awaiting settlement.
No part of this document may be maanual, reproduced or translated into another language without the prior consent of Hewlett-Packard GmbH. Use Power device on Press the green button to get it into a sane mode It will now do mS pulses. The HP A needs time to interpret and implement the commands which it receives.
Transition Controis The transition mode keys enable the operator to select the type of pulse transition and to vary transition time.
During this interim period donations and continued membership are greatly appreciated while we transition to our new space. If a modulated output is required, select the required modulation using the [CTRL] key. Testing Performance Test Equipment Table Double Pulse Adjustment The instrument power fuse is located on the rear panel. E is connected to the micropocessor clock output. Test results recorded at incoming inspection can be used for comparison after carrying out maintenance, repair or adjustments.
The lowest-priced item that has been used or worn previously. For better readability it is recommended to use a space character.
Equipment/HP A Programmable Pulse Generator – London Hackspace Wiki
Width Generator Signal Levels We normally ship within one business day of payment clearing our Pal Pal account. The whole counter counts down when clocked by the burst-clock signal from the main board.
Range switching circuits are included in the outputs of these DACs, that are selected automatically by the microprocessor in conjunction with the Timing range decoder. Errors read with lERR: Comments Hl section fines – If it is not, then the microprocessor is not free running. A continuous pulse train is generated. Selection Pressing a parameter key selects the parameter indicated by the illuminated mnemonic above the key.
Address Decoder Signatures Answer This is the time taken by the HP Ato transmit a message when error reporting or learn mode. Whenever the status byte is read directly after sending a command, the HP A should be given sufficient time to update the status byte.