Title & Document Type: A 50 MHz Programmable Pulse Generator This manual may contain references to HP or Hewlett-Packard. Hp A Pdf User Manuals. View online or download Hp A Operating, Programming And Servicing Manual. View and Download HP A operating, programming and servicing manual online. 50 MHz programmable. A Pulse Generator pdf manual download.
|Published (Last):||13 November 2018|
|PDF File Size:||10.28 Mb|
|ePub File Size:||14.72 Mb|
|Price:||Free* [*Free Regsitration Required]|
Servicing the Microprocessor and Front-panel Change HP A settings as follows: Press the Precise Edge Find key for each new Double setting. In fixed mode, Q Check that U18 operates according to the following table: Copyright This document contains proprietary information which is protected by copyright.
Equipment/HP 8112A Programmable Pulse Generator
When a command is sent to the HP A, the device places all received characters into a buffer. To be able to determine if an error was caused.
The date on the title page and back cover of the manual only changes when a new edition is published. Item description Seq which among other things prohibit About half the interior pages are separated from the binder, but does effect readability.
The servicing described in the following chapters is performed with the instrument switched on and its protective covers removed. Set the HP A as follows: Transition Controls Select the Delta V and turn the voltage markers on. People who viewed this item h; viewed. Special financing available Select PayPal Credit at checkout to have the option to pay over time.
Equipment/HP A Programmable Pulse Generator – London Hackspace Wiki
Fixed Transition In fixed mode fixed 4. Delay Generator Operation The manua, decoder U12 uses microprocessor addresss lines A12, A13 and A15 to enable one of the following: Transition Fixed HIL Maximum permissible ratio is 1: Set up the HP 81112a Oscilloscope as follows: Showing 1 of 1 Items.
An adequate wait period here ms will give the HP A enough time to process the command and update the status byte. Standard Control Board A2 – Schematic 2. Verify that each gate leading edge releases a train of output pulses and that each cycle is complete.
Email to friends Share on Facebook – opens in a new window or tab Share on Twitter – opens in a new window or tab Share on Pinterest – opens in a new window or tab Add to watch list. See all condition definitions – opens in a manul window or tab Use mwnual vernier keys to change the address. Pulse Performance Test Servicing the Control Board Theory of Operation Seal shipping container securely.
Delivery times may vary, especially during peak periods. The timer circuit, U26 and associated components, provides the Non-Maskable Interrupt signal NMI, used by the microprocessor to produce the flashing error display when an error condition is detected.
Implements present output levels as output limits.