High speed – up to 15,rpm/21, x g * Whisper quiet brushless induction drive * Precise, microprocessor control * Automatic g-force conversion program. Hermle ZM-2 Microcentrifuge from Labnet International, Inc. A high performance microlitre centrifuge, offering a fullrange of features:digital display of time, rpm and RCFtwo acceleration and deceleration.
|Country:||Bosnia & Herzegovina|
|Published (Last):||10 August 2017|
|PDF File Size:||20.15 Mb|
|ePub File Size:||5.32 Mb|
|Price:||Free* [*Free Regsitration Required]|
The Hermle Hfrmle M-2 is neither explosion proof nor inert gas shielded and should, therefore, never be operated in explosive-hazardous locations. Never stay in the safety zone of 30 cm around the centrifuge or deposit dangerous materials inside this zone during centrifugation. Commercially available tubes or bottles made of glass or plastic. Width 28 cm Depth 37 cm Height The rotor must always be balanced in this fashion prior to operation. Before operation, secure the rotor lid to the rotor by pressing the snap Before operation, secure the rotor lid to the rotor by pressing the snap Before operation, secure the rotor lid to the rotor by pressing the snap Before operation, secure the rotor lid to the rotor by pressing the snap On herlme digital display “time min-sec” the letters “ro” rotor appears.
Turn the speed knob 1see figure 4 till the required rotor code appears on the d i s p l a y. Operation Operation Operation Operation Operation x233.
The digital indications on the display will light up k-2 the power is on. The green control lamps 3 show which mode is activated. With the centrifuge lid open you can preset the operating time with the knob 3. During a run and with the centrifuge lid closed you have to press the “preset” key 1hold it and change operating time by turning the knob 3.
Figure 8 If the green “fast” control hefmle is lit, the unit accelerates and decelerates fast. If the green “slow”control lamp is lit, the unit accelerates and decelerates very softly below 2 rpm.
Centrifuges Z M-2 -Hermle- Germany
To open the centrifuge lid. The temperature rise depends on z2333 rotor swing-out or angle rotor ,bucket type, ambient temperature, running time and the speed of the rotor.
Use only original spare parts! Should this happen, the centrifuge, rotor, buckets, adapters and the rotor chamber must be thoroughly cleaned and all broken particles removed immediately. If this is not done, they could scratch the protective coating of the rotor.
Hermle LaborTechnik GmbH – Z 233 M-2 Microlitre Centrifuge
There is a distinction between two different kinds of errors. Power failure while rotor is in motion Action: Open and reclose the lid, restart the centrifuge Error no.: Overcurrent or undervoltage due to power supply fluctuations Action Page of 20 Go.
Page 14 – Keyboard – Starting the centrifuge – “qu Page 15 – Temperture Page 16 – Service and inspection of centrifuge Page 17 – Cleaning of centrifuge after breakage of Table of Contents Add to my manuals Add. General Information Hermoe Hermle Z M-2 is neither explosion proof nor inert gas shielded and should, therefore, never be operated in explosive-hazardous locations. Description – The rotor is not installed securely on the motor shaft.
G Hermle Z M-2 Labnet Centrifuge W/rotor | eBay
Set Or Change Rotor-code -Load the rotor symmetrically by inserting tubes into spaces opposite from each other.
Preselection Of Operating Time 4. Keyboard – Starting The Centrifuge – “quick”-key 4. Service And Inspection Of Centrifuge 7.
Page 19 Error no.: Page 20 Hemle no.: Comments to this Manuals Your Name. Enter text from picture: By pressing ‘print’ button you will print only current page. To print the manual completely, please, download it. Don’t show me this message again. Got it, continue to print. Don’t have an account?